It is possible to control a new Huber unit with Pilot ONE controller via a PLC/DCS etc.
Pilot ONE controllers have an Ethernet interface included as standard.
Setting up and using the Ethernet connection:
- To activate the connection: in the Pilot ONE control panel navigate to Menu > Interfaces > Ethernet > Remote Access > Always On.
- Set the Huber unit’s IP address as desired. This setting can be accessed at Menu > Interfaces > Ethernet > IP Address
- Establish a TCP (Transmission Control Protocol) connection to the Huber unit’s IP address and use the TCP port 8101.
- Use the Huber commands:
PB commands (recommended)
- You cannot use the Modbus TCP protocol.
For information on the RS232 and AIF (4-20 mA) options for new Huber units with Pilot ONE, please refer to the separate FAQ entries.